This is a electric monitoring test multimeter which can be used to test power, energy, voltage and current then displayed on a PC or other terminals. It can store data when power off, and store the accumulated energy data before power off. Designed with one key reset button, you can reset energy data by simply pressing one button. The TTL serial port can communicate, read and set parameters with a variety of terminals through a USB adapter.
►This is a electric monitoring test multimeter which can be used to test power, energy, voltage and current.
►The connection of this module is divided into 2 parts: voltage, current input connection and serial port communication, and you can use different TTL adapters to make it communicate with various terminals.
►Comes with a random color current transformer and a USB adapter for convenient use so you do not have to buy separately.
►There is a button for energy data reset, simply press and hold the button for 5 seconds until display flashes and release, then press the button again to finish energy reset.
►It has four digital displays and can also displays through PC or other terminals. You can download the software for PC. For other terminals, it need to be programmed by yourself.
1. Voltage(V), current(A), average power(kW), energy(kWh) measurement.
2. Energy reset by pressing one button.
3. Store data when power off(store the accumulated energy data before power off).
4. LCD display function(display voltage, current, average power, energy).
5. Serial port communication function(TTL serial port can communicate, read and set parameters with a variety of terminals through a USB adapter).
6. Overload alarm.
7. Set power alarm threshold.
Working Voltage: 80~260VAC
Test Voltage: 80~260VAC
Rated Power: 100A/22000W
Working Frequency: 45-65Hz
Measurement Accuracy: ±1.0%
Size: 118 x 65 x 25mm / 4.64 x 2.56 x 0.98"
1. Power Test Range: 0~22kW
0 ~ 10kW, the display format is 0.000 ~ 9.999
10 ~ 22kW, the display format is 10.00 ~ 22.00
2. Energy Test Range: 0~9999kWh
0 ~ 10kWh, the display format is 0.000 ~ 9.999
10 ~ 100kWh, the display format is 10.00 ~ 99.99
100 ~ 1000kWh, the display format is 100.0 ~ 999.9
1000 ~ 9999kWh and above, the display format is 1000 ~ 9999
3. Voltage Test Range: 80~260VAC
Display Format is 110.0 ~ 220.0
4. Current Test Range: 0~100A
Display Format is 00.00 ~ 99.99
Serial Port Communication
This multimeter is design with TTL serial port data communication interface, you can read and set the relevant parameters via serial port. If it need to communicate with a device which has USB or RS232 (such as computer), a TTL adapter is required. In the below table are the communication protocols of this module:
|Voltage||B0||C0 A8 01 01 00 (Computer sends a command to read the voltage value)||1A|
|A0||00 E6 02 00 00 (Meter reply the voltage value is 230.2V)||88|
|Current||B1||C0 A8 01 01 00 (Computer sends a command to read the current value)||1B|
|A1||00 11 20 00 00 (Meter reply the current value is 17.32A)||D2|
|Average Power||B2||C0 A8 01 01 00 (Computer sends a command to read the average power value)||1C|
|A2||08 98 00 00 00 (Meter reply the average power value is 2200w)||42|
|Read Energy||B3||C0 A8 01 01 00 (Computer sends a command to read the energy value)||1D|
|A3||01 86 9f 00 00 (Meter reply the energy value is 99999wh)||C9|
|Set Module Address||B4||C0 A8 01 01 00 (Computer sends a command to set the address, the address is 192.168.1.1)||1E|
|A4||00 00 00 00 00 (Meter reply the address was successfully set)||A4|
|Set Power Alarm Threshold||B5||C0 A8 01 01 14 (computer sends a command to set a power alarm threshold)||33|
|A5||00 00 00 00 00 (Meter reply the power alarm threshold was successfully set)||A5|
Illustration of Communication Protocol:
1. Set the communication address: 192.168.1.1
Send Command: B4 C0 A8 01 01 00 1E
Reply Data: A4 00 00 00 00 00 A4
Note: The above example illustrate that setting the communication address as 192.168.1.1 (the user can set their own address based on their preferences and needs), sending command and automatic replying data are as shown above, the data are expressed in hexadecimal; the last byte of the sending and replying data are 1E and A4, which is cumulative sum. In sending commands: B4 + C0 + A8 + 01 + 01 + 00 = 21E (hexadecimal), the cumulative sum data is 21E, take the last two bytes 1E as the cumulative sum data in sending commands; in reply data: A4 + 00 + 00 + 00 + 00 + 00 = A4 (hexadecimal), the cumulative sum data is A4, which is the cumulative sum data in reply.
2. Set the power alarm threshold: 20 KW
Send command: B5 C0 A8 01 01 14 33
Reply data: A5 00 00 00 00 00 A5
Note: 14 in the sending command is the alarm threshold value (14 is a hexadecimal data representation, which converted to decimal is 20). The power alarm value of this module is based on KW units, and the minimum alarm value is 1KW, the maximum value is 22KW.
3. Read the voltage
Send command: B0 C0 A8 01 01 00 1A
Reply data: A0 00 E6 02 00 00 88
Note: Reply voltage data is D1D2D3 = 00 E6 02, 00 E6 represent the integer-bit of the voltage, 02 represent the decimal of the voltage, the decimal is one digit, converts 00 E6 to decimal is 230; converts 02 to decimal is 2, so the current voltage value is 230.2V.
4. Read the current
Send command: B1 C0 A8 01 01 00 1B
Reply data: A1 00 11 20 00 00 D2
Note: Reply current data is D2D3 = 11 20, 11 represent the integer-bit of the current, 20 represent the decimal of the current, the current decimal is two digits, converts 11 to decimal is 17; converts 20 to decimal is 32, so the current value is 17.32 A.
5. Read the power
Send command: B2 C0 A8 01 01 00 1C
Reply data: A2 08 98 00 00 00 42
Note: Reply power data is D1D2 = 08 98, converts 08 98 to decimal is 2200, so the current voltage value is 2200W.
6. Read the energy
Send command: B3 C0 A8 01 01 00 1D
Reply data: A3 01 86 9F 00 00 C9
Note: Reply energy data is D1D2D3 = 01 86 9F, converts 01 86 9F to decimal is 99999, so the accumulated energy is 99999Wh.
Illustration of Communication
1. Connect hardware according to the wiring diagram below.
2. After wiring, please choose the communication port, this module's host computer software support communication port: COM2COM3COM4, you can check through device manager, if it is not the above communication port, you should amend it.
1.This is an indoor module and can not be used outdoors.
2.Load can be over rated power.
3.Wiring can not be wrong.
1 x Multimeter
1 x USB Adapter
1 x Current Transformer
Free Shipping & Returns
100% Money refund
Fast send and delivery